The position of the metallic circles interior the metallic channel makes exceptionally troublesome their discovery utilizing the non damaging drawn closer over said. In this paper we propose a conceivable modify- local, employing a attractive contact-less estimation method, particularly designed to that point employing a numerical investigation based on the Limited Component Strategy (FEM). A magneto-resistive sensor is connected to the return channel of the ballscrew and the yield signals are appropriately analyzed and post- handled to identify the balls position and movement, in arrange to have parametric data around the ball keenness, channel utilization and other normal blame.
The values of the voltage crests of the sensors are recorded, and their dissemination is utilized to identify atypical condition. Counterfeit blame conditions have been tentatively replicated, and the non dangerous demonstrative framework expectations have been approved by means of a high-speed video checking both for customary working and blame conditions. Post-processed information in time space are utilized to classify harmed and undamaged ballscrews. A few tests have been done, related to diverse setups, and tall estimation repeatability has been found.
The sphere motion and position inside the tube is detected using a permanent magnet and a magnetoresistive sensor. The permanent magnet is positioned near to the tube, so that the spheres are exposed to the magnetic field when they move from the last thread to the first one. The amplitude and the direction of the magnetic field vector change during the sphere transit since the sphere material is in general weakly ferromagnetic. In particular the variation of the magnetic field direction can be detected with high accuracy using a biaxial magnetoresistive sensor based on a Weathstone bridge scheme. This sensor is able to detect the magnetic field angle respect to a reference orientation. The permanent magnet used in this work is a cube of 10 mm of edge, the material is a NdFeB alloy with residual magnetism between 1.29 and 1.32 Tesla, and coercive field strength between 860 and 955 kA/m, the Curie temperature is 310 ?C. The magnetoresistive sensor is a Honeywell? HMC1501 type, that is an integrated small sensor expecially dedicated to linear, angular, or rotary displacement for magnetic field. The sensors utilize the Anisotropic Magnetoresistive (AMR) technology6–8 and its main characteristics are: voltage supply 5 V, angle range from -45 to 45 degrees, sensitivity 2 mV/deg, resolution 0.07 deg, operating temperature from -40 to 125 ?C.
Wanfu Precision Co.,Ltd